Physio-Control notifies AED customers about battery issue
Redmond, WA – Physio-Control has issued a voluntary field action of its LIFEPAK 1000 automated external defibrillator, citing reports of unexpected shutdowns while patients were being treated with the device.
In a Jan. 13 press release, the company states that the “intermittent connection is the result of wear and subsequent oxidation formation between the battery and device electrical contacts.” The company has instructed device owners to immediately remove and reinstall the battery on a weekly basis. Devices exposed to vibration and with a non-rechargeable battery installed for longer periods are more susceptible to the problem.
The company is determining hardware correction changes to eliminate the issue and will contact customers to schedule a device correction when specifics are finalized. Customers will be contacted by mail, but can check updates online.
Customers with questions can call Physio-Control at (866) 231-1220 from 9 a.m. to 7 p.m. Eastern, Monday through Friday or email [email protected].
Post a comment to this article
Safety+Health welcomes comments that promote respectful dialogue. Please stay on topic. Comments that contain personal attacks, profanity or abusive language – or those aggressively promoting products or services – will be removed. We reserve the right to determine which comments violate our comment policy. (Anonymous comments are welcome; merely skip the “name” field in the comment box. An email address is required but will not be included with your comment.)